Midmark Ritter M9 Autoclave
Defect: C660 THROUGH C677 PRESSURE HARDWARE or PRESSURE OVERLIMIT
Chamber pressure is outside the limits for normal operation.
Unplug unit power cord for 1 minute and then plug back in . If problem persists, begin a repair below.
If above solution does not resolve defect, the device will most likely require service or repair.