Midmark Ritter M9 Autoclave
Defect: pressure relief valve seems to be faulty- steam is escaping from valve.
Pressure relief valve not properly seated.
The pressure relief valve may need to be reset per monthly maintenance operations. The relief valve may also be defective or have suffered trauma causing pressure to release at all times- preventing sterilization.
If above solution does not resolve defect, the device will most likely require service or repair.