Initial Inbound Diagnostic Evaluation:
Device has undergone inbound diagnostic evaluation.
Repair / Service Updates:
October 20, 2022
The real-time Dallas clock has reached end of life and needs to be replaced. When the clock is replaced, software values for calibration will need to be restored. This will take about 10 days to get the component here, but can be done no problem.
October 27, 2022
In queue for Circuit Specialist; currently awaiting the RTC battery component
November 9, 2022
November 21, 2022
Repair done. Device is being packaged and readied for return shipment to the customer.
November 23, 2022
Return shipping tracking information:
UPS : 1Z3W9F050393811817
|LINE OPCODE TECH TYPE HOURS||TOTAL|
|Initial Diagnostic Evaluation||0.00|
|[LAB]_002 - Disassembly of device to component level basis||175.00|
|[LAB]_210 - Diagnostic scope of damage/component failure/trace damage/power failure on main logic PCB (motherboard) (Oscilloscope)||85.00|
|[LAB_510] - Repair or correction of damage/component failure/trace damage/power failure on main logic PCB (motherboard) via flux and soldering or other conductive filler||565.00|
|pt_FDT710_realtimeclk_dallas_malaysia-- licensed cal. values added for device recognition||455.00|
|Complimentary Cleaning + Disinfect||0.00|
|Return Shipping / Freight||0.00|
|PLEASE PAY THIS AMOUNT||1,280.00|